There are three items: The cord set is HAR-Certified, and a special mark that will appear on the outer sheath, or on the insulation of one of the inner conductors. Refer all servicing to qualified service personnel. To prevent shock, connect the medical monitor to personal computer that complied with this standard. The wires in this mains lead are coloured in accordance with the following code:
|Date Added:||7 October 2014|
|File Size:||46.70 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
There are three options for the automatic timeout: Always use the appropriate AC cord that is certified for the individual country. Cet appareil numerique de la Classe B respecte toutes les exigences du Reglement sur le materielle brouilleur du Canada.
MQ Touch Signage Monitor
These two merits make it easy to use in almost any environment and easy to transport. We recommend that you also consult your Graphics card User’s manual during the Troubleshooting Procedure. Video Analog RGB 0.
Cautions and Warnings Table of Contents 1. If you find that any of these items are damaged or missing, please contact your dealer immediately. This removes dust and other particles. Make sure your graphics adapter has the following settings: Connect the signal cable on the rear of the monitor to the pin connector on the rear of the computer.
To unlock, do the same procedure again.
Power off both the Computer and Monitor before making any connections. The graphics card must also be set multi correctly in Windows and make sure the video output of the graphics card is listed in Section 6.
Appendix FOR medical model only: Install Monitor on a solid horizontal surface such as a table or desk. A slider with mu,tiq value is displayed.
MultiQ 219, User Guide
Shielded interconnected cables and shield power cords must be employed with this equipment to insure compliance with the pertinent RFD emission limits governing this device. Do not install this monitor near a motor or transformer where strong magnetism is generated.
Use a power cable that is properly grounded. The Screen Adjustment z 8. Windows drivers for both the monitor and the graphics card are supplied on the Windows CD or diskettes. This is used myltiq adjust the sound volume. Some examples are listed below: After the question listed above is solved, we continue the setup procedure as per below.
MQ219 Touch Signage Monitor
It may not function properly if any of the following applies: The adjustable items will be displayed in the main menu. Hot air may cause damage to the cabinet and other parts. If this equipment does cause harmful interference to radio or television reception, which can be determined by turning the equipment off and on, the user is encouraged to try to correct the interference by one or more of the following measure: You may use the standard resolution SXGA x 60 as the monitor type.
Operation is subject to the following two conditions: To prevent shock, connect the medical monitor to personal computer that complied with this standard. Please consult the table below for the selection criteria for power cords used in the United States and Canada. This device must be operated with the original power supply, part number: Push the MENU mhltiq once to display the main menu of the screen adjustment.